Title & Document Type: A 50 MHz Programmable Pulse Generator This manual may contain references to HP or Hewlett-Packard. Hp A Pdf User Manuals. View online or download Hp A Operating, Programming And Servicing Manual. View and Download HP A operating, programming and servicing manual online. 50 MHz programmable. A Pulse Generator pdf manual download.
|Genre:||Health and Food|
|Published (Last):||18 September 2004|
|PDF File Size:||14.84 Mb|
|ePub File Size:||3.49 Mb|
|Price:||Free* [*Free Regsitration Required]|
The burst flip-flop is located on the main board, not on the control board. Hpp item – opens in a new window or tab. This item may be a floor model or store return that has been used. The signal passes through a divider circuits before being applied to the VCO.
HP 8112A Manuals
About Manial Test Solutions Marvin Test Solutions, a vertically-integrated aerospace test and measurement company, has created and delivered innovative, reliable test systems for factory, depot, intermediate, and flightline use since See other items More Select each output parameter in turn by pr essing its asso ciated key. Test Preperetipn Refer to Chapter Connect the HP A and counter as shown in Figure Figure and Figure show the front and rear panel respectively. Digital to Analog Converters, timer, reference circuits, and the width-vernier current source.
This question appears to be off-topic. In DBL mode the width generator is configured to produce an output pulse at both leading and trailing edges of the trigger pulse from the delay generator. Display shows max value available in selected range.
The decoder output lines determine which timing, mode or range circuit latches data from the data bus LDO to LD7 as shown in Table In the “Width and delay loop” lines toevery 2 second delay is incremented by 1 ms while width is decremented by 2 ms. Microprocessor board architecture Recommended Test Equipment 3- 1. Adjustment Procedures Back to home page Return to top. Timing 1C biock diagram Refer to Figure This input is used to initialize the GPIA.
Full text of “HP: A”
Counter reset For the counter to be re-enabled, the logic signal LBC from the address decoder U16 must be set true low. Adjust the parameter value using the f vernier! TRE 10 NS 4. If a cable is to be locally manufactured, use male connector, HP part number No part of this document may be photocopied, reproduced or hhp into another language without the prior consent of Hewlett-Packard GmbH.
Use a 50 D feedthrough termination 0.
Inspect the shipping container for damage. Verify that each external trigger pulse generates a burst of 50 output pulses as shown here.
HP A Compatible Pulse Generator – GPH | Marvin Test Solutions, Inc.
Typical signals in Gate mode To prevent potentiai fire 88112a shock hazard, do not expose the HP A to rain or other excessive moisture. External Trigger Controls The limit 1 key LED is lit when this mode is active.
This input works in conjunction with a bias adjusting network and the internal manuxl shift, to provide normal and complement output modes. Check that all boards are properly fitted. Status Reporting Structures are required. The microprocessor scans the frontpanel key matrix using a BCD to decimal converter U20 and an 8-to-l multiplexer U