Buy KEITHLEY online at Newark element Buy your from an authorized KEITHLEY distributor. The Model Electrometer combines flexible interfacing capabilities with current sensitivity, charge measurement capabilities, resolution, and speed that are. Product Features: Programmable electrometer; Less than 1fA noise; >TW input impedance on voltage measurements; Charge measurements from 10fC to .
|Published (Last):||18 March 2005|
|PDF File Size:||18.76 Mb|
|ePub File Size:||9.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
Low Voltage Burden The ‘s feedback ammeter design minimizes voltage offsets in the input circuitry, which can affect current measurement accuracy. Built-in IEEE and RS interfaces make it possible to program all instrument functions over the bus through a computer controller.
Model front face. Bulk capacitors are listed in table below: Was it worth doing?
Any error signals that remain are negligible in comparison keithle those that can occur when measuring current with a DMM. DMMs measure current using a shunt resistance that develops a voltage typically mV full-range in the input circuit. Instrument have no missing parts and overall very clean. Model Image 6: As with all recent Keithley gear, schematics or detailed construction information is not available.
| Keithley Electrometer repair
Model Image 8: Main PCBtop and bottom view Image 5: Performance test … Conclusion …. They usually have voltage, current, resistance and charge measurement functionality, just like usual DMMbut can do these measurements with minimal loading on the circuit under test. The ‘s feedback ammeter design minimizes voltage offsets in the input circuitry, which can affect current keithlry accuracy. In addition to use by researchers in areas such as physics, optics, and materials science, the ‘s affordable price makes it an attractive alternative to high end DMMs for low current measurement applications, such as testing resistance and leakage current in switches, relays, and other components.
Electrometers reduce this terminal voltage by using the feedback ammeter configuration.
Providing the finest test equipment solutions since 1992
Redistribution of article must retain the above copyright notice, this list of conditions, link to this page https: Voltage Burden and Measurement Error Electrometers provide current measurement with lower terminal voltage than is possible when making DMM measurements.
Many other Keithley Instruments firmware dumps are available on xDevs.
Model front face Image 2: Main PCBtop and bottom view. If you find typo or have any questions feel free to leave comment or contact us.
Economical Component Testing Once, keihhley were simply considered too slow to keep up with the high throughput that production test applications demand.
These instruments usually have special low-leakage shielded input connectors, often with triaxial design with PTFE insulators, Image 1: Typically, offset currents in DMMs are tens or hundreds of picoamps, which severely limits their low current measuring capabilities compared to the with 3fA input bias current. The Keithley Programmable Electrometer combines flexible interfacing capabilities with current sensitivity, charge measurement capabilities, resolution, and speed that are equal or superior to our earlier electrometers.
High resistivity measurements Leakage currents Ion selective electrode measurements pH measurements Conductivity cells Potentiometry. Redistribution and use of this article or any files referenced in it, in source and binary forms, with or without modification, are permitted provided that the following conditions are met:.
Keithley Programmable Electrometer- at the Test Equipment Depot
Time worklog summary Total man-hours spent on this project around 1 hoursgive or take few. Data Sheet KB. This leakage current represents an error keithlry. Two analog outputs-a 2V output and a preamp output-are available for recording data with stripchart recorders.
The lowers this terminal voltage still further-to the level of the voltage noise-by canceling out the small keityley voltage that remains. Resistor network TF used to generate secondary voltages and currents.
Model Image 9: Electrometers provide current measurement with lower terminal voltage than is possible when making DMM measurements. The first current is the dark current I D generated by the detector with no light falling upon the device in other words, the signal of interest ; the second one is the leakage current I Keifhley generated by the voltage burden VBURDEN appearing at the terminals of the ammeter.